- User Since
- Aug 1 2016, 12:10 PM (123 w, 1 d)
Thu, Nov 29
Started implementation in the branch, but doesn't work yet. Benjamin, can you fix this at the community day?
I think the main problem is the 3D rendering update. With the suggestions written down in the last comments we can however reach about 20 fps which is ok for the demos. We wont fix this now.
Nov 4 2018
Nov 3 2018
Pushed new branch T25608-SimpleOpenCVExample.
Nov 2 2018
A possible workaround is to use PLUS for the connection to the Epiphan grabber and then connect to it via OpenIGTLink:
(1) Start a PLUS server (2.6.0 / 64bit was tested) with the default device set for a Epiphan frame grabber (PlusDeviceSet_Server_EpiphanVideoCapture.xml)
(2) Open MITK, create a new video device in the US view and choose "OIGTL Client".
(3) Activate the device and it should work
Tested the setup with two Epiphan framegrabbers of the same type and DVI/VGA video sources. The problem is always there. A OpenCV minimal example works fine, but if we paste exactly the same code to a MITK plugin the problem is there. Seems as if OpenCV doesn't like MITK running in the background for this specific frame grabber. With a framegrabber from another manufacturer (TERRATEC G1) or with a webcam video source there was no problem during the tests.
Oct 30 2018
Oct 13 2018
Pushed new branch T25494-InstallerCleanup-not-for-master.
Oct 12 2018
Oct 8 2018
The final version implemented by the students is available here: https://github.com/Alfred-Franz/MITK/tree/T24520-AblationPlanning
Oct 5 2018
Oct 4 2018
Pushed new branch T25494-CleanupHummelProtocolMeasurements.
Sep 28 2018
Pushed new branch T25478-RenderingManagerContinousUpdate.
I also tried to move the update code of the US plugin to a thread, but it seems as if the USDevice class is not thread safe.
Sep 27 2018
Moved update code of tracking toolbox to a thread. Works for the moment, with the following problems:
- Seems as if the TrackingToolStatusWidget doesn't like if it's updated from another thread. Some buttons in the same UI part are blinking from time to time. If we don't update the widget from the thread there is no problem but we don't see the tool status.
- Updating the color of the tool itself (green for valid / red for invalid) is not done at the moment, because some information is missing in the thread. Do we really need this feature?
- Update rate is constant for now. The update loop in the thread can wait a certain time if we hand over a update rate from the UI.
Pushed new branch T25478-MoveUpdateCodeToThreads.
Pushed new branch T25477-Add-StandardICP.
Sep 6 2018
Pushed new branch T24287-v2018.04-beta-integration-2018-09.
Sep 5 2018
I just tried to build the current beta branch with Ubuntu 18.04 and NavigationModules configuration. I merged this branch, but the build failed. It seems OpenIGTLink was correctly downloaded by the superbuild, but later this error occurred:
Aug 10 2018
Aug 8 2018
Aug 7 2018
Pushed new branch T25227-SimplifyToolMetaDataHandling.
Pushed new branch T25224-SpacingCalibration.
Pushed new branch T25223-FreezeAbstractTrackerDevice.
Pushed new branch T24287-v2018.04-beta-integration.
Aug 6 2018
Pushed new branch T24287-Integration-IGTRelease-2018-03.